Neonatal intubation is an essential but difficult skill to learn. Videolaryngoscopy allows the airway view to be shared by the intubator and supervisor and improves intubation success. Ideally, a videolaryngoscope (VL) should be usable as a conventional laryngoscope (CL). The aims of this report were to describe differences in the shape of currently available CL and VL blades and to compare the direct airway view obtainable on a neonatal manikin with different laryngoscope blades.
Three main differences were observed; compared with CL, the VL blades have a reduced vertical height, a curved tip and curved body. The direct airway view obtained by the VL is narrower than that obtained with the CL, although the corresponding view on the VL screen is maintained.
Minor adaptation of intubation technique may be required when using a VL. Modifying VL blades to reduce these differences may improve their usefulness as an intubation training tool.
- endotracheal intubation
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.
Contributors Both authors contributed equally to the design, undertaking, writing and editing of this report.
Funding The authors have not declared a specific grant for this research from any funding agency in the public, commercial or not-for-profit sectors.
Competing interests None declared.
Provenance and peer review Not commissioned; externally peer reviewed.
Data sharing statement No unpublished data available from this study.